A High Voltage (HV) pulser to produce very short pulses, at high voltage, using only a 5V input and 3.3 to 5V logic to control the duration of the pulses (from 20ns to 150ns+), based on the HV7360.
The High Voltage level can be set up between 30V to 120V.
For ultrasound processing, there is a MD0100 clipping the signal coming back from the transducer to a +-2V.
The pins layout is as follows:
I made it to play with ultrasound transducers. I needed a module to manage the HV excitation for my transducer.
Use of a HV7360 pulser and a HighVoltage recom : makes is robust and low-power
|Shipping Rate||First item||Additional items|
La Poste: Slower tracked parcel
We recognize our top users by making them a Tindarian. Tindarians have access to secret & unreleased features.
We look for the most active & best members of the Tindie community, and invite them to join. There isn't a selection process or form to fill out. The only way to become a Tindarian is by being a nice & active member of the Tindie community!
Was a former ultrasound engineer.
Co-founder of echOpen's project at echopen.org
The work on Tindie is strictly my own.