A current shunt for directly displaying current on a regular voltage channel of an oscilloscopeDesigned by LeoNerd's Store in United Kingdom
Still active despite the COVID-19 situation; though shipping times will be slightly slower at the moment.
New version 0.6 with plastic BNC connectors, reducing the weight. Some older photos still show the older metal version - replacement photos will be available soon Summary Convert current to voltag...Read More…
New version 0.6 with plastic BNC connectors, reducing the weight. Some older photos still show the older metal version - replacement photos will be available soon
It's basically a small series resistor acting as a current shunt with a current-sense amplifier across it. Connect the PSU to the Vin side of the probe, the device under test to the Vout side, and the probe will convert the amount of current drawn by the device into an output voltage on the I[out] BNC socket, proportional to the current drawn at 10V of output per 1A of current draw. Just configure the channel as a "x0.1 probe", and the values given by the channel now show the current in amps.
It's aimed at low-power devices, intended largely for running 5V or 3.3V supplies, up to a few hundred mA in current. The shunt resistor has a value of 0.1 ohms, so even at 200mA it drops only 20mV; your 5V supply will still be usable at 4.98V while even 3.3V remains good at 3.28V. You could even use the Vout monitor on the sense input of your PSU if it has one, to let it account for this voltage drop itself, and ensure your load really does receive the exact voltage required.
There are two options to power the sense amp. It can be powered by the input supply so it does not need a separate PSU, though this does limit the available voltage swing on the I[out] monitor output. Or, you can supply a separate source of power for the amplifier than is powering the load; either to give the current output signal a greater range, or because you want to power the load at less than 3V.
With the maximum current, keep in mind that while the device itself can handle up to 1.5A though it, it won't be able to indicate this as a voltage on the I[out] socket unless it is supplied by at least 15V. Conversely, if powered by only 3V, it will only be able to indicate a current draw of 300mA. The device under test may draw more than this, though it will not be possible to directly measure how much using the output.
The shunt resistor is placed across the positive supply to measure high-side current, leaving the ground a low-impedance path between the supply and the load. Additionally this is the ground that the output is referenced to. This means you can happily power your device from a ground-referenced power supply if required, without upsetting the oscilloscope. Also you can use other oscilloscope probes in different parts of the circuit, because they all share a common ground. If you wish to power the sense amplifier with its own supply, it must use a common ground with this.
To connect the monitor outputs from this probe to your oscilloscope you will need to use two BNC-to-BNC patch cables of some kind. I have a handy pair of such cables in nicely-flexible RG174 with coloured identification markers for sale - Twin BNC RG174 Patch.
You may also find it useful to have a "Look Hook" Power Cable to connect the output to a board under construction, as these probes allow connection to an unpopulated through-hole on a PCB, such as for a pin header or power socket.
Alternatively, the Pogo Pin Power Cable makes a convenient temporary connection to an unpopulated 2-pin 0.1" header footprint.
Current probe adapters are generally available from some of the higher-end oscilloscope manufacturers, but they tend to be very expensive, carrying a 4-figure price tag because they're capable of measuring a huge range of very small to very large currents, at possibly hundreds of volts, at bandwidths going into the 100s of MHz. They also tend to require a separate power supply costing up to that much again.
By comparison, this probe is aimed more towards the hobbyist end of the spectrum. It is much cheaper though obviously its specifications won't fully match these kinds of numbers. The current-sense amplifier on this board is rated up to 28V, and has a quoted bandwidth of 1.8MHz - still ample to view current consumption transients at the microsecond scale. Input bias noise within the sense amplifier will limit its accuracy at tiny currents below 1mA or so, and the maximum amount of current it can handle is 1.5A.
This product is mainly aimed at USB-powered devices, or at least those that would fit within an envelope of 5V at 500mA. As such, it has a USB miniB socket to connect to the supply, and a USB A socket to connect to the load device. There's also a pair of 4mm binding posts on either side, as well as 0.1" pitch pins and sockets; meaning there's many possibilities to connect supply and load to the probe. In addition, the D+ and D- lines of the USB sockets are connected together, meaning that USB communication through the probe is not affected; it should be totally transparent to USB data flow. Finally, an additional space is reserved for a 3pin connector carrying ground, Iout and Vout signals, should you wish to measure the output via something other than the BNC connectors. These holes are left unpopulated, allowing you to put in a header, pin socket or other connector as you require.
No country selected, please select your country to see shipping options.
No rates are available for shipping to .
Enter your email address if you'd like to be notified when Oscilloscope Current Probe Adapter can be shipped to you:
Thanks! We'll let you know when the seller adds shipping rates for your country.
|Shipping Rate||Tracked||Ships From||First Item||Additional Items|
Alex | Jan. 1, 2017
Josh | Jan. 21, 2016
We recognize our top users by making them a Tindarian. Tindarians have access to secret & unreleased features.
We look for the most active & best members of the Tindie community, and invite them to join. There isn't a selection process or form to fill out. The only way to become a Tindarian is by being a nice & active member of the Tindie community!