Find broken coax cables using this tester and your oscilloscopeDesigned by land-boards, Ships from United States of America
What is it? Fast Edge Pulse Generator for use as a Time Domain Reflectometer (TDR). Used to measure cable length and test continuity/termination of 50 or 75 Ohm coax cables. 3D Printed case in pict...Read More…
Fast Edge Pulse Generator for use as a Time Domain Reflectometer (TDR). Used to measure cable length and test continuity/termination of 50 or 75 Ohm coax cables.
3D Printed case in picture not included - 3D printer files are here.
Requires a fast oscilloscope for precision measurement. To use it, plug one BNC using male-to-male adapter into a scope and attach a cable to the other BNC connector. Allows measurement of the cable and determination if there is a cable defect (open/short, termination, Etc.).
This video, by W2AEW, inspired me to want to make a Pulse Generator for use with an oscilloscope as a Time Domain Reflectometer (TDR). There were some DIY boards available but none that are as good as this design.
Within a month of first building this board, I had a bad piece of RG-8X coax and it was difficult to diagnose the problem. I used an ohmmeter to verify that the coax was good from a DC perspective and it looked fine while I was testing with the meter. It was connected end to end and there were no opens or shorts that I could see with the meter.
However, I knew the cable was bad because I swapped it out for another cable and the other cable worked just fine. When I went to use the broken cable, I could jiggle the cable and make it work and then it would flake out but it was far from consistent. Intermittent problems are the worst sort of problems, aren't they? As they say - Murphy's Law is a universal constant.
I decided to hook the questionable coax up to this Pulse Generator board and the results were impressive to say the least. When I flexed the cable the waveform went from a nice stepped waveform to nearly flat on my oscilloscope (a little bump which went down quickly). That told me that the problem was on the near end of the cable and that it was a short between the center conductor and the shield. In the end the problem was a short at the PL-259 connector which (once found) was easily repaired.
|Shipping Rate||First item||Additional items|
United States Postal Service: First Class
United States Postal Service: Priority Mail (3 Days) w tracking
Bill | Feb. 15, 2016
Garnet | March 9, 2015
Vincent | Feb. 17, 2015
We recognize our top users by making them a Tindarian. Tindarians have access to secret & unreleased features.
We look for the most active & best members of the Tindie community, and invite them to join. There isn't a selection process or form to fill out. The only way to become a Tindarian is by being a nice & active member of the Tindie community!
I have been designing electronics circuit cards for over 30 years with many successful card designs. Check out my webpage to see the sort of cards I have designed and built. I also have a BS in Electrical Engineer and a Master's Degree in Computer Engineering from USC.
My three sons and I run land-boards to make the sort of boards that we find useful. My oldest son is an experienced programmer and is a Python wizard. He writes our test code. My middle son works professionally as an electronics assembler and he is working his way towards getting an Electrical Engineering degree. My younger son is only in High School but he's a very good computer programmer as well.
We do good stuff, together.