Find broken coax cables using this tester and your oscilloscopeDesigned by Land Boards, LLC in United States of America
Wiki Page for the Fast Pulse Generator What is it? Fast Edge Pulse Generator for use as a Time Domain Reflectometer (TDR). Used to measure cable length and test continuity/termination of 50 or 75 O...Read More…
Fast Edge Pulse Generator for use as a Time Domain Reflectometer (TDR). Used to measure cable length and test continuity/termination of 50 or 75 Ohm coax cables.
3D Printed case in picture not included - 3D printer files are here.
Requires a fast oscilloscope for precision measurement. To use it, plug one BNC using male-to-male adapter into a scope and attach a cable to the other BNC connector. Allows measurement of the cable and determination if there is a cable defect (open/short, termination, Etc.).
This video, by W2AEW, inspired me to want to make a Pulse Generator for use with an oscilloscope as a Time Domain Reflectometer (TDR). There were some DIY boards available but none that are as good as this design.
Within a month of first building this board, I had a bad piece of RG-8X coax and it was difficult to diagnose the problem. I used an ohmmeter to verify that the coax was good from a DC perspective and it looked fine while I was testing with the meter. It was connected end to end and there were no opens or shorts that I could see with the meter.
However, I knew the cable was bad because I swapped it out for another cable and the other cable worked just fine. When I went to use the broken cable, I could jiggle the cable and make it work and then it would flake out but it was far from consistent. Intermittent problems are the worst sort of problems, aren't they? As they say - Murphy's Law is a universal constant.
I decided to hook the questionable coax up to this Pulse Generator board and the results were impressive to say the least. When I flexed the cable the waveform went from a nice stepped waveform to nearly flat on my oscilloscope (a little bump which went down quickly). That told me that the problem was on the near end of the cable and that it was a short between the center conductor and the shield. In the end the problem was a short at the PL-259 connector which (once found) was easily repaired.
No country selected, please select your country to see shipping options.
No rates are available for shipping to .
Enter your email address if you'd like to be notified when Coax Tester - Fast Edge Pulse Gen. (50 or 75 Ohm) can be shipped to you:
Thanks! We'll let you know when the seller adds shipping rates for your country.
|Shipping Rate||Tracked||Ships From||First Item||Additional Items|
We keep a small quantity of assembled/tested boards but we have material in-house to build/test additional boards. Domestic (USA) orders ship as soon as the cards is built/tested. We typically build larger orders and ship on weekends.
Bill | Feb. 15, 2016
Garnet | March 9, 2015
Vincent | Feb. 17, 2015
We recognize our top users by making them a Tindarian. Tindarians have access to secret & unreleased features.
We look for the most active & best members of the Tindie community, and invite them to join. There isn't a selection process or form to fill out. The only way to become a Tindarian is by being a nice & active member of the Tindie community!