DC Electronic Load Tester 35W 5A LCD Display Adjustable Constant Current Aging Resistor Discharger Voltage Current Power Battery CapacitySold By ICStation, Ships from China
Express Shipping (DHL/FedEx) can be offered for order reached $200. If you need it, please leave a message while ordering. Thanks for your support!
1.Description: The industrial-grade long-life electronic load tester adopts long-life ball fan which has fast rotating speed and strong wind.It supports continuous discharging for non-stop work a...Read More…
The industrial-grade long-life electronic load tester adopts long-life ball fan which has fast rotating speed and strong wind.It supports continuous discharging for non-stop work and suitable for long-period discharge or the factory do aging test for products. It is suitable for testing the charger's output current capability, charger batch aging test, measuring battery load capacity, etc.
It can be used as ordinary electronic load, also do USB electronic load use, support QC2.0/QC3.0/AFC/FCP Fast charge protocol, trigger output 5v/9v/12v/20v and other voltages, the use of liquid crystal display, friendly human-computer interface and excellent performance indicators, meet a wide range of testing needs.
Automatically display the last parameter value.
It can be set that KZ35 can be set manually or automatically restore normal operation (REC) after the load is released from the protection mechanism. - Status ON:The load automatically restore normal operation after the protection mechanism is released. - Status OFF:The load needs to be manually restored to normal operation after the protection mechanism is released.
AUTO function is used to set whether automatically trigger the fast charge function after power ON. - Status ON:Automatically trigger the fast charge function after power ON.It is suitable for batch testing aging charger. - Status OFF:Does not automatically trigger fast charge function.
A1:Connect the load, short press the trigger button ‘TRG’, enter the fast charge protocol selection interface.Select QC2.0 mode by rotary potentiometer.Short press the potentiometer.LCD will dynamically display ‘----’.The second line display '-5.0' after activate QC2.0 fast charge mode.Then rotating potentiometer to select '-12.0'.The first line will show the actual voltage.Check if the actual voltage is about 12.0V.And then press potentiometer return to the load interface.Press button ‘STOP’ to turn the load and the slowly adjust the current to 2A.Check if the voltage has a significant reduction (>0.5V).Then let the load work for a period of time, and measure the heat of the charger. If the load restarts or the voltage reduction significantly or the charger heats up sharply, it means that the charger load capacity is not sufficient and must reduce current to keep normal use.
Q2:Is my PD2.0 charger supported?
A2:The load can do aging test for the PD charger, but PD seduction is not possible. The two-headed type-c line connects the PD charger and the load, and the PD charger outputs 5V by default. If the PD charger supports QC2.0 at the same time.The load can do aging test for the PD charger.
Q3:I want to release fast charge, how to achieve?
We recognize our top users by making them a Tindarian. Tindarians have access to secret & unreleased features.
We look for the most active & best members of the Tindie community, and invite them to join. There isn't a selection process or form to fill out. The only way to become a Tindarian is by being a nice & active member of the Tindie community!