Extender/risers boards for repair & troubleshooting of HP(Agilent) Test Gear including 5345A frequency counter and similar. FREE SHIPPING!Designed by IR AnalysIR-Infrared decoder & anlayzer OCXO 10Mhz in Ireland
What is it? We have sets of extender/riser boards or cards for older HP (Agilent) test gear. Designed originally for the 5345A frequency counter, the boards will also fit many other HP systems from...Read More…
We have sets of extender/riser boards or cards for older HP (Agilent) test gear. Designed originally for the 5345A frequency counter, the boards will also fit many other HP systems from the same era, without modification. Specifically, 20P (2x10) and 44P (2x22) connectors are available by default. With care, smaller configurations can be achieved by removing (cutting) part of the lower connector to fit and aligning the target card to the remaining edge of the socket.
For use with the 5345A, 2x20P and/or 2x44P cards are required. In the 5345A there are typically 6 or 7 cards with 2x44P connectors and 3 cards with 2x20P connectors. The 2 power supply cards have 2x20P connectors each. If you know which board is faulty, then you only need to purchase 2 for that card. The cards are raised by just over 100mm (~4 inches). If you prefer additional height / boards clearance, then order double the quantity above for double the height etc. The board thickness is 1.6mm, which fits snuggly into the existing sockets. Because of the use of solder-mask on both faces of the boards, there is little chance of accidental 'shorts' unlike the original HP riser boards.
The boards come with 20P or 44P card connectors pre-soldered. The width of the green 20P card is 43.1mm, the green 44P card is 90.6mm. The blue card edge connector is slightly larger, as per photos.(20P:47mm, 44P:94.5-108mm approx.), The critical dimension is the card width, as these standard 3.96mm pitch connectors are designed to fit these widths.
If you are unsure if these cards will work with other models, please contact us to discuss.
Now with FREE SHIPPING worldwide!
A common story: bought an old 5345A online as 100% working, but it had a fault on delivery. The only way to repair & troubleshoot test equipment from this era is by using extender cards like this. Unfortunately, these cards are very hard to find at any price.
It is possible, to make these cards yourself using acid etching of copper PCB boards, cut to the correct dimensions. However, I prefer to avoid etching and toxic fumes if possible, which comes with the benefit of higher quality results using our boards.
These high quality extenders cards are very rare, and difficult to find individually or as a complete set for the 5345A and similar test gear.
As a bonus, you should be able to resell these boards online after you are finished using them.
Select one of the options described here, over on the right:
If you have other sizes you would like to see available on Tindie, let us know the details and target device.
Delivery is via standard post worldwide and usually takes from 5->10 working days after shipping. Standard post is untracked and without insurance. We do not cover any customs or import tax charges, which if levied are the responsibility of the buyer.
|Shipping Rate||Ships From||First item||Additional items|
An Post: FREE Standard Post
Gordon | July 7, 2018
We recognize our top users by making them a Tindarian. Tindarians have access to secret & unreleased features.
We look for the most active & best members of the Tindie community, and invite them to join. There isn't a selection process or form to fill out. The only way to become a Tindarian is by being a nice & active member of the Tindie community!